FAQ and knowledge base
Information about what data is shared with videometer when enabling telemetry can be found here.
Help in the User Manual
If an issue occurs with your equipment, please follow the directions in the VideometerLab software under Help -> User Manual.
Q1) I am not able to acquire images all the time or sometimes
I get error messages such as: Capture Error: GrabberReturnedError_MissingPackets, Missing frames
For JAI camera (series 3 or earlier), ensure 'GigE Network adapter setup' in GigE JAI Camera Setup.pdf has been followed. We recomend the following power configuration for stable image acqusition: Windows Power Settings 1.0.pdf
Read about our instrument-telemetry policy (information shared with videometer to improve our software): Instrument-telemetry
Q2) I need support with my software since operation 'X' fails
A2.1) In case an algorithm does not work as expected, you are welcome to contact us regarding the issue, in most cases this can be fixed either by another algorithm or by tuning the parameters
A2.2) For errors a dialog should appear to transmit error data to our bugtracking system, if the error is critical for daily usage then please write to us and we will prioritize according to the service contract associated with the instrument. NB - If the instrument PC is not connected to the internet, it is required to send the data manually, this can in VideometerLab versions 3.12.x and newer be done using the "Help->System footprint" following the guide.
Q3) The VideometerLab software could not be installed
Please confirm that you have administrator rights in the VideometerLab pc. Then check the disk space as the installation and Instrument Calibration requires minimum 1 GB of free disk space.
Please make sure that there is no old version of the software installed or running at the installation pc. You may remove any VideometerLab by selecting:
- "Start" -> "Settings" -> "Conrol Panel" -> "Add or Remove Programs"
- Try the installation again
Q4) The light does not flash but one color is constantly on
This light is normally a result of a communication error between the pc and the VideometerLab. Please follw the description in the VideometerLab software under Help -> User Manual.
Q5) Why is the spere not completely dark in the stand-by mode?
Some diodes may send out light when the instrument is powered but the VideometerLab application is not running. This is expected and ensures stable temperature of the LED's.
Q6) The VideometerLab cannot be calibrated
If it is the first time the VideometerLab is calibrated, please check that you have at least 1 GB free disk space on the drive where the software is installed - usually the C drive.
If the VideometerLab software reports an error, please make sure that the calibration plates are used in the correct order and that they are clean and without any damages.
Follow the description in the VideometerLab software under Help -> User Manual. If the problem persists, please report the problem to us.